I am preparing a smart wearable medical product for the elderly.I have some questions about ADPD188GG, and I look forward to your answers. I will to test the CNiBP, SpO2 and HRM / HRV of the elderly.so,
1. CNiBP : AD8223+ADPD188GG, connected to PD1…
I am new with ADPD188GG PPG Sensor and I need to know the specification of the maximum peak current of the VLED to test it with its Evaluation board in the Float mode.
I would to know the number of pulse in both modes (Float mode & normal mode)??
What is the difference between the ADPD188BI and the ADPD188GG used for PPG measurements?
It should be supported based on the fact that there is accelerometer on the board. I personally do not have experience with ADPD188GG evaluation board, but hardware sync should be supported based on the devices on the board.
The initial check I would…
i have a question, about ADPD188GG....
i see the spec ADPD188GG has internal 2 Green LED (TX) and 2 PDET (RX) with IR cut filter... is right??
and then it has external 2 LED driver and 2 PDs Input ... is right???
does it mean i can…
My client want to evaluate the ADPD188GG for PPG solution.Do you have any ME portion design guide of ADI?
If the device is operating in a single timeslot and configured to write a 16 bit result to the FIFO per sample then 2 bytes would be written to the FIFO. If the device is operating with two timeslots and using both PD channels with 32 bits per…
According the datasheet VLED1(pin 5) voltage range is 4.0~5.0V. The max rating of VLED1 is 5V. If we provide 5V supply and 5% tolerance
The max supply voltage is 5.25V. Is VLED1 can endure this application?
I'm trying to create a driver for the ADPD188GG, with the float mode configuration (ADPD188GGZ_PPG_Float_01.dcfg)
I used the HAL you shared on github).
So I made these calls to a initialize the AFE,(the example adpd188gg)